LargeV Dental X-ray Digital Tomography Equipment (HiRes3D CBCT),
namely SmartV. This software is an associated software for the
LargeV HiRes3D Dental X-ray Digital Tomography Equipment and its
purpose is to function in conjunction with the system for the
maxillofacial (including the temporomandibular joint, TMJ) 3D
photography and image processing.
1. Patient Scan Module
The Interface of Scan Module This interface includes entry bar for
patient information, setting bar for scanning parameters, display
bar for equipment status, progress bar for scanning course and
display window for real-time projection.
2. Image Processing Module
The software will switch automatically to this module interface
once a record in the data
module is double clicked (or the label is clicked), and the
corresponding image of the record will be displayed. The module
interface includes four parts: main toolbar, page switching bar,
image display field and image processing control panel.
3. Patient Data Management Module
Shortcut Toolbar Shortcut toolbar is applicable for fast data
import/export, inquiry of patient record and other functions.
Record List Bar This area displays the patient record imported to
the system or the corresponding inquiry result of patient record.
Record Detail Bar Record detail bar displays the detail information
including image serial number, amount of image, type of imaging,
build time and relevant depictions of selected patient. Image
Preview Window Image preview window displays the axial plane image
of selected patient and provides corresponding preview.
4. Report Module
A. Report toolbar :The report toolbar provides report printing,
report saving, report layout, report editing, cloud report
configuration, page creation/deletion and other tools.
B. Report editing window: User can perform report layout, editing
and other operations in the report editing window.
C. Image display bar :The screenshot/original image sent from the
image page will be displayed in the image display bar and user can
select or delete the images needed in report.
D. Report management bar: The report management bar displays the
catalog of reports saved in native machine and servers for the
viewing convenience of users.
|Field of View|
(mm * mm)
|160 x 150|
160 x 80
80 x 80
50 x 80
|230 x 180|
160 x 100
160 x 50
80 x 80
|Voxel Size (mm)||0.25|
|Best Spatial Resolution||2.6 lp/mm||2.4 lp/mm|
(Min / Max)
|≤30s / 150s||≤40s / 60s|
|Max Exposure Power||600W|
|Scan Time||15s, 18s, 21s||15s|
|Exposure Time||13s, 16s, 19s||13s|
|Focal Spot Size||0.5(IEC336)|
|Sensor Type||CsI+CMOS Flat Panel Detector||CsI+α-Si Flat Panel Detector|
|Power Requirements||AC220V±10%, 50Hz, 2000VA|
|Outer Dimensions||1077mm x 1499mm x 2109mm|
|Room Requirements||1.5m x 1.8m x 2.5m|
|* The data are subject to change without notice.|